Publication: High resolution simultaneous dual liquid level measurement system with CMOS camera and FPGA hardware processor
Full text at PDC
Advisors (or tutors)
Elsevier Science BV
In this paper a new high precision automatic measurement system for liquid level measurement in membrane distillation application is presented. Membrane distillation (MD) is a relatively new process that is being investigated worldwide. It has some advantages in relation to other separation processes such as conventional distillation and reverse osmosis (110), which are: high quality of distillate, low operating temperatures, low operating pressure, low power consumption allowing the use of alternative energy sources such as solar and geothermal energy.For MD process it is necessary a very high precision system to measure the solution level in the reservoirs, able to detect changes smaller than 0.05 mm. This is due to the small fluxes through the membrane, and therefore it is necessary to detect the small changes produced in the liquid levels in the reservoirs. Until now such measurement has been made manually, by means of a cathetometer, but it resulted very uncomfortable. In order to overcome this issue, a high resolution and accuracy system (0.013 +/- 0.008 mm), that makes automatic measurements of both reservoirs, cold and hot, is proposed. This system is based on a laser beam, assisted with a high resolution 3 Mpixels CMOS camera and a computer vision processor implemented in a FPGA, with a measurement rate up to 12 times per second.
© 2013 Elsevier B.V.
 K. W. Lawson, D.R. Lloyd, Review membrane distillation, J. Membr. Sci. 124(1997) 1.  M.A. Izquierdo Gil, M.C. García Payo, C. Fernández Pineda, Direct contactmembrane distillation of sugar aqueous solutions, Sep. Sci. Technol. 34 (9) (1999) 1773.  J. I. Mengual, J.M. Ortiz de Zárate, L. Peña, A. Velázquez, Osmotic distillation through porous hydrophobic membranes, J. Membr. Sci. 82 (1993) 129.  M. Khayet, P. Godino, J.I. Mengual, Nature of flow on sweeping gas membranedistillation, J. Membr. Sci. 170 (2000) 243.  M.A. Izquierdo Gil, G. Jonsson, Factors affecting flux and ethanol separation performance in vacuum membrane distillation (VMD), J. Membr. Sci. 214(2003) 113.  M.A. Izquierdo Gil, M.C. García Payo, C. Fernández Pineda, Air gap membrane distillation of sugar aqueous solutions, J. Membr. Sci. 155 (1999) 291.  M.C. García Payo, M.A. Izquierdo Gil, C. Fernández Pineda, Air gap membrane distillation of aqueous alcohol solutions, J. Membr. Sci. 169 (2000) 61.  Model 2700 Multimeter/Switch System User’s Manual, Keithley Instruments,Inc., Cleveland, OH, U.S.A., August 2003, Seventh Printing, Document Number:2700-900-01 Rev. G.  C.-C. Hsu, M.-C. Lu, W.-Y. Wang, Y.-Y. Lu, Distance measurement based on pixelvariation of CCD images, ISA Trans. 48 (October (4)) (2009) 389–395.  F. Lucklum, B. Jakoby, Non-contact liquid level measurement with electromagnetic–acoustic resonator sensors, Meas. Sci. Technol. 20 (12)(2009) 124002 (7pp).  R. Casanella, O. Casas, R. Pallàs Areny, Continuous liquid level measurement using a linear electrode array, Meas. Sci. Technol. 18 (7) (2007) 1859–1866.  C. Vázquez, A.B. Gonzalo, S. Vargas, J. Montalvo, Multi-sensor system using plastic optical fibers for intrinsically safe level measurements, Sens. ActuatorsA: Phys. 116 (October (1)) (2004) 22–32.  H. Canbolat, A novel level measurement technique using three capacitive sensors for liquids, IEEE Trans. Instrum. Meas. 58 (October (10)) (2009) 3762–3768.  M.-C. Lu, W.-Y. Wang, H.-H. Lan, Image-based height measuring systems for liquid or particles in tanks, in: 2004 IEEE International Conference on Networking,Sensing and Control, Vol. 1, 21–23 March, 2004, pp. 24–29.  E. Fysikopuolos, M. Georgiou, N. Efthimiou, S. David, G. Loudos, G. Matsopoulos,A Spartan3E-based low cost system for gamma-ray detection in small single photon emission computed tomography or positron emission tomography systems, Meas. Sci. Technol. 22 (11) (2011) 114002 (7pp).  S. Strube, Gabor Molnar and Hans-Ulrich Danzebrink Compact field programmable gate array (FPGA)-based multi-axial interferometer for simultaneous tilt and distance measurement in the sub-nanometre range, Meas. Sci.Technol. 22 (9) (2011) 094026 (8pp).  M.G. Lorenz, L. Mengibar, E. SanMillán, L. Entrena, Low power data processing system with self-reconfigurable architecture, J. Syst. Archit. 53 (9) (2007)568–576.  M.G. Lorenz, M.A. Izquierdo Gil, R. Sánchez Reillo, C. Fernández Pineda, Single CMOS sensor system for high resolution double volume measurement appliedto membrane distillation system, Proc. SPIE 6491 (2007) 64910P.  Xilinx Inc., Spartan-3 Starter Kit Board User Guide, 2013,http://www.xilinx.com (retrieved June 2013).  Inc. Micron Technology, MT9T001 – 1/2-Inch 3-Megapixel Digital Image Sensor,2013, http://www.micron.com (retrieved Jun 2013).  Maxim Integrated Products MAX5230/MAX5231 12bits dual DACs Datasheet,2013, http://www.maxim-ic.com (retrieved June 2013).  M.A. Izquierdo Gil, C. Fernández Pineda, M.G. Lorenz, Flow rate influence ondirect contact membrane distillation experiments: different empirical correla tions for Nusselt number, J. Membr. Sci. 321 (2008) 356–363.